ISO 20776 PDF
ISO. First edition. Clinical laboratory testing and in vitro diagnostic test systems — Susceptibility testing of infectious agents and evaluation. ISO () Clinical laboratory testing and in vitro diagnostic test systems – Susceptibility testing of infectious agents and evaluation of performance of. Buy ISO , Susceptibility testing of infectious agents and evaluation of performance of antimicrobial susceptibility test devices, Part 1, at CLSI.
|Published (Last):||10 October 2009|
|PDF File Size:||2.21 Mb|
|ePub File Size:||9.28 Mb|
|Price:||Free* [*Free Regsitration Required]|
Contact Sitemap Privacy Statement Disclaimer. In summary, our data suggest that microdilution trays for hydrophilic drugs such as FLC can be accurately prepared following the ISO recommendations or the serial 2-fold dilution schemes.
Simultaneous quantification of voriconazole and posaconazole in human plasma by high-performance liquid chromatography with ultra-violet detection. 207776 control procedures of AST should be implemented in order to minimize the inaccuracy of dilution methods and lack of sio when hydrophobic compounds such as ITC are tested.
Open in a separate window. Calibration and validation – archive In order to determine and compare the accuracy of each method described for microdilution, two consecutive HPLC determinations of the azole concentration were calculated from different wells of trays prepared according ISO recommendations, serial dilutions for hydrophobic drugs, and serial dilutions for hydrophilic compounds. However, itraconazole trays showed a significant lack of reproducibility that was independent of which method was followed.
Where clinical data is lacking! Development and validation of a high-performance liquid chromatographic assay for the determination of fluconazole in human whole blood using solid phase extraction. Susceptibility testing of infectious agents and evaluation of performance of antimicrobial susceptibility testing devices.
EUCAST: External documents
Author information Article notes Copyright and License information Disclaimer. These documents are available for purchase from ISO or national standards organisations. Fluconazole trays can be accurately prepared by following ISO and serial dilution schemes. Sitemap Search close sitemap. This azole compound can only be ionized and solubilized in water at a very low pH, and also earlier studies have indicated that poor solubilization can have a significant influence on generated MIC values Furthermore, previous studies have demonstrated a ieo higher interlaboratory variation in MIC results for quality control strains for itraconazole than for flucytosine and fluconazole 6 oso 8.
According to those oso and the ISO recommendations, the 2-fold dilutions of compounds should be performed in a multistep, rather complex, and time-consuming fashion, avoiding serial dilutions in order to prevent the risk of a systematic error being induced, for instance, by an imprecise pipette 9. Susceptibility testing of yeasts Considering only data for dilution series prepared in Spain, no systematic difference was detected between ISO dilution and serial dilution. Reference method for testing the in vitro activity of antimicrobial agents against rapidly growing anaerobic bacteria involved in infectious diseases.
Since the hydrophilic growth medium RPMI-G at 2776 7 has been selected for testing in vitro antifungal susceptibility, it was important to describe the lack of proper solubility of this compound.
General Committee to Journal List J Clin Microbiol v. Selective high-performance liquid chromatographic assay for itraconazole 2076 hydroxyitraconazole in plasma from human immunodeficiency virus-infected patients.
FLC eluted at 4. Clinical breakpoints for antimycobacterial agents Prevalidation statistical design to assess analytical methods.
First indications of improved capability and capacity of public health microbiology laboratories in the European Union http: ISO documents of relevance to antimicrobial susceptibility testing These documents are available for purchase from ISO or national standards organisations. ECDC roadmap for integration of molecular typing and genomic typing into European-level surveillance and epidemic preparedness — Version 2. Fifty microliters of the resultant mix was injected directly into the HPLC system.
Relevant external documents
Stock solutions and microdilution wells were tested for concentration of antifungal agent using high-performance liquid chromatography HPLC methods previously described 125. Meetings, Minutes and Reports However, preparation of 207766 trays for antimicrobial susceptibility testing AST microdilution tests is associated with a significant lack of reproducibility, particularly after freezing and shipping, independent of which dilution method was followed.
Splitting wild type distributions.
Received Jan 20; Accepted Feb Sensitive determination of itraconazole and its 207776 metabolite in human plasma by column-switching high-performance liquid chromatography with ultraviolet detection. Bioanalytical method validation-a revisit with a decade of progress. Previous versions of documents AST of mycobacteria Reference method for testing the in vitro activity of antimicrobial agents against rapidly growing aerobic bacteria involved in infectious diseases.
Rapid AST in bloodcultures