HACH SC100 PDF
This item is no longer available. Suggested replacements. UVAS sc Probe, 1 mm, with SC Controller · Get Quote. Continuous UV Results 1 – 20 of Search results for sc Amtax sc Ammonium Analyzer, mg/L NH₄-N, V, 5 m Filterprobe sc. This item is no longer available. Suggested replacements. UVAS sc Sensor, 5 mm, with SC Controller · Get Quote. Continuously protect plant treatment.
|Published (Last):||11 September 2013|
|PDF File Size:||11.96 Mb|
|ePub File Size:||13.2 Mb|
|Price:||Free* [*Free Regsitration Required]|
The calibration history log shows the gain value, the time and date of the calibration, and the initials of the operator performing verification. Section 4 Operation 4.
Hach-Lange SC User Manual | 58 pages
Enter text from picture: Refer to Figure 29 and perform the following steps to replace the fuses: In addition, the passcode also regulates function selection for relay options. Page 71 Index Sample Line Installation Section 7 Troubleshooting 7. Remove the quick connect fitting and wires and install the threaded plug on the opening to maintain the environmental rating. Maximum cable length is 9. Troubleshooting Troubleshooting Table 11 presents sensor warnings displayed in the Event Log, possible causes, and corrective actions.
Instrument Setup Operation 4. Pass the cable through conduit and a conduit hub or a strain relief fitting and an available access hole in the controller enclosure. Changing The Sensor Name Operation 4. Calibrate Sensor as required by regulatory agency Per agency-dictated schedule. Section 10 Repair Service Authorization must be obtained from Hach Company before sending any items for repair.
Don’t have an account? The time is available only in hour military format.
Hach sc100 1720E Instrument Manual
Inadequate bubble removal from Increase the sc00 averaging time to a longer interval. Replacing The Lamp Assembly Section 4 Operation Section 4 Operation 4.
Troubleshooting Section 7 Troubleshooting Specifying The Displayed Language Operation 4. Wear cotton gloves to protect your hands and to avoid fingerprints on the bulb. Section 7 Troubleshooting Section 7 Troubleshooting 7. Troubleshooting Table 11 presents sensor warnings displayed in the Event Log, possible causes, and corrective actions.
By pressing ‘print’ button you will print only current page. Restore sample flow to the instrument. Section 2 General Information 2. Index Index Numerics 90 Degree Detector Rinse with deionized water before calibrating.
Hach-Lange SC 100 User Manual
After all plumbing and electrical connections have been completed and checked, place the head on the body and supply power to the system. Don’t show me this message again.
Page 39 Operation 4. Table of Contents Section 1 Specifications Vis spectrophotometer for hach and hach lange water analysis methods 24 pages. Tug gently to check the connection.
Page of 72 Go.
Setting Up System Security Operation 4. Installation Electrical Installation High-voltage wiring for the controller is conducted behind the high voltage DANGER The instrument must be installed barrier in the controller enclosure.
Section 11 Limited Warranty In the event that a defect is discovered during the warranty period, Hach Company agrees that, at its option, it will repair or replace the defective product or refund the purchase price, subject to the pro-rated schedule above, excluding original shipping and handling charges.
In the event that a defect is discovered during the warranty period, Hach Company agrees that, at its option, it will repair or replace the defective product or refund the purchase price, subject to the pro-rated schedule above, excluding original shipping and handling charges.
Page 35 Operation 4. Allow at least mm Sensor Setup Menu Operation 4. Table of Contents Section 5 System Startup Please contact the Hach Service Center serving your location.
Section 1 Specifications Specifications are subject to change without notice. Wet Verification System Startup 5. Measured reading based on a gain of 1. The barrier must remain in place unless a by qualified technical personnel qualified installation technician is installing wiring for power, alarms, or relays.
To locate the representative nearest you, send e-mail to intl hach. Table 1 Turbidimeter Specifications Nephelometric light scatter at 90 degrees relative to the incident light beam.