EXFO AXS-110 MANUAL PDF
EXFO’s AXS Handheld OTDR series covers the whole range of short-haul OTDR test The AXS All-Fiber OTDR combines singlemode and multimode . Search among more than user manuals and view them online EXFO’s AXS does not interfere with the CO’s laser transmitters, as it uses. otherwise, without the prior written permission of EXFO Electro-Optical. Engineering Inc. Use the following general instructions to navigate through the AXS/ menus and Specify a value between 0 and dBrnC if the ANSI.
|Country:||Papua New Guinea|
|Published (Last):||19 January 2011|
|PDF File Size:||17.10 Mb|
|ePub File Size:||20.99 Mb|
|Price:||Free* [*Free Regsitration Required]|
To acquire traces in Manual Advanced mode: Page 46 Testing Fibers Testing in Auto Mode Testing in Auto Mode The application will automatically evaluate the best settings according to the fiber link currently connected to the unit in less than 5 seconds.
Page 99 11 Inspecting Fibers with the FIP The fiber inspection probe FIP allows you to find dirty or damaged connectors by displaying an enlarged view of the connector surface.
Page 91 Measuring Power or Loss Nulling Offsets Nulling Offsets Temperature and humidity variations affect the performance of electronic circuits and optical detectors, which can offset measurement results. Page 5 Contents 12 Testing Network Connections By default, fiber characteristics are evaluated each time you start a test.
EXFO AXS/ Ethernet Test Set User Manual | pages
Does not include uncertainty due to fiber index. For loss measurements, reference your power meter to axd-110 light source see Referencing Your Power Meter to a Manjal on page 84and then deactivate the light source. If you select the Bellcore. After saving a result, the unit prepares the next file name by incrementing the suffix. In-Stock Notification Request Sent Your request to be notified when this product is back in stock has been sent to our sales department.
Page Troubleshooting Transportation You may also be requested to provide the firmware or software version numbers. Please contact us via email or toll free at and we will be happy to assist you with this request. For the most recent version of this spec sheet, please go to the EXFO website at http: You should perform a shutdown if you do not intend to use your unit for a week or more. Mouse move on Image to zoom. You can also include information about the tested fiber and job or add comments.
Press Enter to open the Distance unit list. To enable or disable the first connector check: When the unit is turned off, it keeps the following parameters in its internal memory: Press Menu, select Power Meter, and then press Enter. You can also modify the time zone. Being manuall not to touch the soft end of the swab, moisten a cleaning tip with only one drop of isopropyl alcohol. When the backlight is dimmed, the unit operation is not interrupted. Answer Yes to confirm.
To select a new interface language: If you want the unit to reset the settings for each acquisition, select No. It verifies the injection level and the reflectance of the first connector.
Repeat steps 3 to 4 with a dry cleaning tip. To clean detector ports: You can set the trace display to show only the portion of the trace that is located between the span start and the span end, in full-trace view.
Index probe autonaming image files Find the folder or disk: Remarks Nominal launch conditions are overfilled.
On the unit keypad, press the? Measuring Power or Loss Referencing Your Power Meter to a Source Referencing Your Power Meter to a Source In reference mode, your unit displays the loss created by the fiber under test only, since it subtracts a reference value from the measured power. Contents 12 Testing Network Connections To turn the unit on: In-service OTDR troubleshooting of optical fiber does not interfere with the normal operation and expected performance of the information channels.
What’s missing? Tell us about it.
Page Maintenance Replacing Batteries Replacing Batteries As soon as you remove the battery compartment door, the unit will turn itself off. Use the arrows to select the Pulse list and press Enter to open the list.
Select duration of the acquisition as follows: